B-Alert in AcqKnowledge 4.4.1 and earlier: Attempting to record data after disabling the “Use B-Alert Definition File” option generates an “acquisition cannot continue” error message.
Note that this issue is common only to AcqKnowledge versions 4.4.1 and earlier.
This issue can occur under the following circumstances:
Launch AcqKnowledge and create a B-Alert graph.
Select B-Alert > Set Up Data Acquisition > Length/Rate and verify that a definition file has been assigned.
Uncheck the “Use B-Alert Definition File” box and close the acquisition setup dialog.
A “Brain state definition file is required for one of the selected channels” message appears. Dismiss this dialog by clicking OK.
Click the Start button.
A second message appears: “There is an error with calculation channel C0. The acquisition cannot continue.”
This error occurs because when a definition file is assigned, calculation channels C0-C7 are automatically enabled to accommodate the Brain State Gauge data. If the “Use B-Alert definition file” is subsequently disabled, these calculation channels remain active even though the definition file is no longer being referenced. If this issue occurs, the following work-arounds are available:
A. Enable the “Use B-Alert Definition File” option and rerun the acquisition.
B. If the definition file is not to be used, disable calculation channels C0-C7 prior to recording:
Select B-Alert > Set Up Data Acquisition > Channels and click the “Calculation” tab.
Uncheck the “Acquire” box for channels C0-C7 and rerun the acquisition.
fNIR Imaging Systems are stand-alone functional brain imaging solutions for continuous NIR spectroscopy (NIRS). They provide an in-lab cognitive function assessment and eliminate a great many of the drawbacks of a functional MRI. fNIR optical imaging technology measures neural activity and hemodynamic response in the prefrontal cortex. Subjects wear an fNIR sensor (IR light sources […]
BIOPAC provides software and hardware that allow for research teams to record and analyze respiration activity in physiological experimentation. Here are a few notable studies covering HRV, EMG, and EEG measurement. HRV (Jing Wei, Hong Luo, Si J. Wu, Paul P. Zheng, Genyue Fu, and Kang Lee) Humans encounter various stressful situations everyday at work, home, and school. Such stress when experienced at high degrees and/or […]
Alan Macy Answers Proust Questionnaire BIOPAC co-founder and R&D Director, Alan Macy, was interviewed recently by Roger Durling for the Santa Barbara Independent. The article, “The S.B. Questionnaire: Alan Macy,” provided a background excerpt of the Santa Barbara entrepreneurs’ past as well as his answers to the Proust Questionnaire. The biomedical technology developed at BIOPAC reflects an ambitious […]