B-Alert in AcqKnowledge 4.4.1 and earlier: Attempting to record data after disabling the “Use B-Alert Definition File” option generates an “acquisition cannot continue” error message.
Note that this issue is common only to AcqKnowledge versions 4.4.1 and earlier.
This issue can occur under the following circumstances:
Launch AcqKnowledge and create a B-Alert graph.
Select B-Alert > Set Up Data Acquisition > Length/Rate and verify that a definition file has been assigned.
Uncheck the “Use B-Alert Definition File” box and close the acquisition setup dialog.
A “Brain state definition file is required for one of the selected channels” message appears. Dismiss this dialog by clicking OK.
Click the Start button.
A second message appears: “There is an error with calculation channel C0. The acquisition cannot continue.”
This error occurs because when a definition file is assigned, calculation channels C0-C7 are automatically enabled to accommodate the Brain State Gauge data. If the “Use B-Alert definition file” is subsequently disabled, these calculation channels remain active even though the definition file is no longer being referenced. If this issue occurs, the following work-arounds are available:
A. Enable the “Use B-Alert Definition File” option and rerun the acquisition. B. If the definition file is not to be used, disable calculation channels C0-C7 prior to recording:
Select B-Alert > Set Up Data Acquisition > Channels and click the “Calculation” tab.
Uncheck the “Acquire” box for channels C0-C7 and rerun the acquisition.
This research system ships with an MP36R Data Acquisition unit and AcqKnowledge software licensed for Workflow Automation. MP36R four-channel data acquisition unit has built-in universal amplifiers that can record a wide range of physiological signals. AcqKnowledge Acquisition & Analysis Software is an interactive, intuitive program that lets you instantly view, measure, transform, and analyze data. Perform complex […]
AcqKnowledge is an interactive, versatile software program that allows you to manage data in real time. Perform complex data acquisition, stimulation, triggering and analyses using intuitive navigation. Here are some recent AcqKnowledge citations… The nature of vicarious embarrassment. Journal of Business Research, 153, 355-364. Ziegler, A. H., Allen, A. M., Peloza, J., & Norris, J. I. (2022). […]
BIOPAC’s comprehensive Introductory ECG Guide addresses fundamental to advanced concerns to optimize electrocardiography data recording and analysis. Topics include: ECG Complex; Electrical and Mechanical Sequence of a Heartbeat; Systole and Diastole; Configurations for Lead I, Lead II, Lead III, 6-lead ECG, 12-lead ECG, precordial leads; Ventricular Late Potentials (VLPs); ECG Measurement Tools; Automated Analysis Routines for extracting, […]