B-Alert in AcqKnowledge 4.4.1 and earlier: Brain State Gauge Surface and Heat Maps are unresponsive during acquisition; colors remain a uniform green (Surface Map) and yellow (Heat Map).
This is most likely because the “No output into ABM format” option is selected in the B-Alert Data Acquisition Settings. In order for Surface and Heat Map output to be recorded, a valid ABM data file output location must be defined.
Note that this issue is common only to AcqKnowledge versions 4.4.1 and earlier. In AcqKnowledge versions 4.4.2 and higher, Heat and Surface Map output will function properly with the “No output” option selected.
Use the following steps to ensure proper Surface and Heat Map operation:
Choose B-Alert menu > Set Up Data Acquisition > Length/Rate.
Click the “Choose ABM data file location” button.
Select the “Output data into ABM data file at:” option, assign a valid file path, and click OK.
Rerun the B-Alert acquisition. The Surface and Heat Map gauges should display the expected color output.
Offering the industry’s only system that delivers a comprehensive, 24-hour data logging solution in an easy-to-use, easy-to-wear package, the BioNomadix Logger® truly delivers “physiology anywhere.” The BioNomadix Logger is the perfect tool for applications that demand greater degrees of subject freedom and complex experimental design. BioNomadix Loggers wirelessly record physiological data as subjects freely and naturally […]
Recent physiological studies feature BIOPAC’s Wireless BioNomadix Data Recorders Using Metronomes and Rhythmic stimulation to help people with Parkinson’s disease walk For those living with Parkinson’s disease, even simple tasks can take too long to accomplish properly, such as crossing a room or picking up an object. In this study, participants with diagnosed cases of […]
BIOPAC’s just released Introductory ECG Guide addresses fundamental to advanced concerns to optimize electrocardiography data recording and analysis. Topics include: ECG Complex; Electrical and Mechanical Sequence of a Heartbeat; Systole and Diastole; Configurations for Lead I, Lead II, Lead III, 6-lead ECG, 12-lead ECG, precordial leads; Ventricular Late Potentials (VLPs); ECG Measurement Tools; Automated Analysis Routines […]