B-Alert in AcqKnowledge 4.4.1 and earlier: Brain State Gauge Surface and Heat Maps are unresponsive during acquisition; colors remain a uniform green (Surface Map) and yellow (Heat Map).
This is most likely because the “No output into ABM format” option is selected in the B-Alert Data Acquisition Settings. In order for Surface and Heat Map output to be recorded, a valid ABM data file output location must be defined.
Note that this issue is common only to AcqKnowledge versions 4.4.1 and earlier. In AcqKnowledge versions 4.4.2 and higher, Heat and Surface Map output will function properly with the “No output” option selected.
Use the following steps to ensure proper Surface and Heat Map operation:
Choose B-Alert menu > Set Up Data Acquisition > Length/Rate.
Click the “Choose ABM data file location” button.
Select the “Output data into ABM data file at:” option, assign a valid file path, and click OK.
Rerun the B-Alert acquisition. The Surface and Heat Map gauges should display the expected color output.
Offering the industry’s only system that delivers a comprehensive, 24-hour data logging solution in an easy-to-use, easy-to-wear package, the BioNomadix Logger® truly delivers “physiology anywhere.” The BioNomadix Logger is the perfect tool for applications that demand greater degrees of subject freedom and complex experimental design. BioNomadix Loggers wirelessly record physiological data as subjects freely and naturally […]
Solving Physical Stress Issues for Surgeons Physical stress in the operating room is common for surgeons, considering how much time they spend in an upright standing position during surgical procedures. With procedures that require standing for several hours and average reported working times of 60-80 hours a week, surgeons seem to be at risk for musculoskeletal […]
BIOPAC combines eye tracking with mobile data logging and physiology data collection to run experiments in the lab or in the real world BIOPAC Systems, Inc. is pleased to announce new eye tracking integration for researchers who conduct experiments from mobile participants in diverse locations. ETVision (EYE-ETV) eye tracking glasses now integrate with BIOPAC wireless systems and software, […]