B-Alert in AcqKnowledge 4.4.1 and earlier: Brain State Gauge Surface and Heat Maps are unresponsive during acquisition; colors remain a uniform green (Surface Map) and yellow (Heat Map).
This is most likely because the “No output into ABM format” option is selected in the B-Alert Data Acquisition Settings. In order for Surface and Heat Map output to be recorded, a valid ABM data file output location must be defined.
Note that this issue is common only to AcqKnowledge versions 4.4.1 and earlier. In AcqKnowledge versions 4.4.2 and higher, Heat and Surface Map output will function properly with the “No output” option selected.
Use the following steps to ensure proper Surface and Heat Map operation:
Choose B-Alert menu > Set Up Data Acquisition > Length/Rate.
Click the “Choose ABM data file location” button.
Select the “Output data into ABM data file at:” option, assign a valid file path, and click OK.
Rerun the B-Alert acquisition. The Surface and Heat Map gauges should display the expected color output.
The fEMG Electromyography Smart Amplifier is designed specifically for recording facial EMG signals and electrical activity from other small muscle groups. Use with AcqKnowledge™ software to analyze facial expressions and startle paradigms. BIOPAC’s new Smart Amplifiers are designed for great data. Smart Amplifiers improve performance by amplifying the physiological signal close to the subject, which allows a high-level […]
BIOPAC provides software and hardware that allows research teams to run autonomic function tests while collecting physiological data. Here are a few studies focusing on BIOPAC’s ability to record physiological signals during autonomic function testing. The Internet, Sleep, and Heart Rate Does internet usage lead to poorer sleep? Using a BIOPAC Data Acquisition System to measure cardiovascular […]
BIOPAC’s comprehensive Introductory ECG Guide addresses fundamental to advanced concerns to optimize electrocardiography data recording and analysis. Topics include: ECG Complex; Electrical and Mechanical Sequence of a Heartbeat; Systole and Diastole; Configurations for Lead I, Lead II, Lead III, 6-lead ECG, 12-lead ECG, precordial leads; Ventricular Late Potentials (VLPs); ECG Measurement Tools; Automated Analysis Routines for extracting, […]