B-Alert in AcqKnowledge 4.4.1 and earlier: Brain State Gauge Surface and Heat Maps are unresponsive during acquisition; colors remain a uniform green (Surface Map) and yellow (Heat Map).
This is most likely because the “No output into ABM format” option is selected in the B-Alert Data Acquisition Settings. In order for Surface and Heat Map output to be recorded, a valid ABM data file output location must be defined.
Note that this issue is common only to AcqKnowledge versions 4.4.1 and earlier. In AcqKnowledge versions 4.4.2 and higher, Heat and Surface Map output will function properly with the “No output” option selected.
Use the following steps to ensure proper Surface and Heat Map operation:
Choose B-Alert menu > Set Up Data Acquisition > Length/Rate.
Click the “Choose ABM data file location” button.
Select the “Output data into ABM data file at:” option, assign a valid file path, and click OK.
Rerun the B-Alert acquisition. The Surface and Heat Map gauges should display the expected color output.
MP160 data acquisition and analysis systems with AcqKnowledge software provide a flexible tool for life science research. MP160 data acquisition workstations include all hardware, software, cables, and guides you need to get started. MP160 Data Acquisition Hardware—Ethernet-ready data acquisition and analysis unit can record multiple channels with differing sample rates; record at speeds up to 400 kHz (aggregate). […]
BIOPAC provides software and hardware that allows research teams to study stimulus presentation data. Here are a few notable studies using BIOPAC equipment for research on stimuli’s effects using ECG and SCL data. Your Face Scares Me This study addressed the impact of individual differences in social anxiety by examining the effects of perceptual load and stimulus valence when […]
BIOPAC’s comprehensive Introductory ECG Guide addresses fundamental to advanced concerns to optimize electrocardiography data recording and analysis. Topics include: ECG Complex; Electrical and Mechanical Sequence of a Heartbeat; Systole and Diastole; Configurations for Lead I, Lead II, Lead III, 6-lead ECG, 12-lead ECG, precordial leads; Ventricular Late Potentials (VLPs); ECG Measurement Tools; Automated Analysis Routines for extracting, […]