When data acquisition is configured to acquire using “Averaging”, some signal amplitude excursions may need to be excluded during the averaging process. Artifact Rejection, available via a checkbox in the averaging setup, is a feature for just this purpose.
To perform on-line averaging, first choose “MP150 > Set Up Acquisition…” (older versions of AcqKnowledge) or “MP150 > Set Up Data Acquisition…” and choose “Length/Rate” in the pane on the left (AcqKnowledge version 4.4). In the top row of pulldown menus, choose “Record” and “Save one” using “Averaging”. Underneath “Averaging”, a “Setup” button will be enabled. Click that button to set up various aspects of averaging (e.g., the number of trials) and click the “Artifact rejection” checkbox.
When artifact rejection is enabled this way, the MP150 System will ignore any trials that contain signals exceeding the artifact rejection thresholds, keep track of how many trials have been rejected, and add that number of trials to the total number of trials to be acquired. This allows users to “re-try” a trial that was rejected due to the presence of artifact.
Workflow—A Great Tool for Every Researcher! Automate AcqKnowledge with New Workflow Tool New automation tool significantly saves time and improves consistency with standardized procedures Workflow’s drag-and-drop interface allows you to automate and repeat the signal-conditioning and processing steps your protocol requires in an easy-to-use process without programming or scripting knowledge. Batch process multiple files with confidence […]
AcqKnowledge is an interactive, versatile software program that allows you to manage data in real time. Perform complex data acquisition, stimulation, triggering and analyses using intuitive navigation. Here are some recent AcqKnowledge citations… The nature of vicarious embarrassment. Journal of Business Research, 153, 355-364. Ziegler, A. H., Allen, A. M., Peloza, J., & Norris, J. I. (2022). […]
BIOPAC’s comprehensive Introductory ECG Guide addresses fundamental to advanced concerns to optimize electrocardiography data recording and analysis. Topics include: ECG Complex; Electrical and Mechanical Sequence of a Heartbeat; Systole and Diastole; Configurations for Lead I, Lead II, Lead III, 6-lead ECG, 12-lead ECG, precordial leads; Ventricular Late Potentials (VLPs); ECG Measurement Tools; Automated Analysis Routines for extracting, […]