This application note details hardware and software setup to synchroniize a B-Alert System with an MP150 System running on two separate computers.
Connect the 3.5 mm phone jack connector to the Analog Output (A0) port on the front of the UIM100C, and attach to the ECG input of the B-Alert Systems.
Set up the MP150 System and AcqKnowledge software as required for the experiment.
In the AcqKnowledge software on the computer for the MP150: Choose MP150 > Set Up Stimulator and set parameters
In the AcqKnowledge software on the computer for the B-Alert: Choose B-Alert > Set Up Channels and check Acquire and Plot for Analog Channel 1 (A1/EKG).
Click the Start button in the AcqKnowledgeSoftware for the B-Alert System.
IMPORTANT: You must start AcqKnowledge for the B-Alert System before starting AcqKnowledge for the MP150 system.
Click the Start button in the AcqKnowledge software for the MP150 System.
Observe the B-Alert data. A negative and positive spike*should appear. (If necessary for a better view, autoscale CH1 and hide additional channels.)
* Due to filtering applied by the EKG channel, the square wave will appear truncated in the B-Alert waveform. If desired, an AC-coupled sine wave can be used instead for a more accurate representation.
After data recording is complete, return to the beginning of the B-Alert graph where the stimulus occurred.
Highlight all data before the stimulus, and choose Edit > Clear All.
The MP150 Data is now synched to the B-Alert Data (time point ‘zero’ of the MP150 data file equals time point ‘zero’ of the B-Alert data file).
Basic principles of physiological data collection covers fundamentals of data collection and equipment configuration equipment to help you refine experiment protocols and avoid costly missteps. Frazer Findlay, CEO of BIOPAC, discusses common mistakes with physiology recording and shares his secrets for collecting great data. Topics include: How to prep a subject and where to place electrodes; Which type of electrodes work for different body signals; Analog to Digital Conversion; Sample rates (correct vs. incorrect); Filtering; and Scaling and calibration.Watch On Demand Now!
Recent psychophysiological studies feature BIOPAC’s MP Research series of Data Recorders and AcqKnowledge software. Creating Adaptive Assistants for helicopter crews based on Real-time Physiological Data In the Mercedes S-Class, biometric tools track a driver’s eyes, heart rate, and other physiological data, with one goal in mind: Making sure drivers are alert and safe on the […]
BIOPAC’s just released Introductory ECG Guide addresses fundamental to advanced concerns to optimize electrocardiography data recording and analysis. Topics include: ECG Complex; Electrical and Mechanical Sequence of a Heartbeat; Systole and Diastole; Configurations for Lead I, Lead II, Lead III, 6-lead ECG, 12-lead ECG, precordial leads; Ventricular Late Potentials (VLPs); ECG Measurement Tools; Automated Analysis Routines […]