This application note details hardware and software setup to synchronize a B-Alert System with an MP150 System running on two separate computers.
Connect the 3.5 mm phone jack connector to the Analog Output (A0) port on the front of the UIM100C, and attach to the ECG input of the B-Alert Systems.
Set up the MP150 System and AcqKnowledge software as required for the experiment.
In the AcqKnowledge software on the computer for the MP150: Choose MP150 > Set Up Stimulator and set parameters
In the AcqKnowledge software on the computer for the B-Alert: Choose B-Alert > Set Up Channels and check Acquire and Plot for Analog Channel 1 (A1/EKG).
Click the Start button in the AcqKnowledgeSoftware for the B-Alert System.
IMPORTANT: You must start AcqKnowledge for the B-Alert System before starting AcqKnowledge for the MP150 system.
Click the Start button in the AcqKnowledge software for the MP150 System.
Observe the B-Alert data. A negative and positive spike*should appear. (If necessary for a better view, autoscale CH1 and hide additional channels.)
* Due to filtering applied by the EKG channel, the square wave will appear truncated in the B-Alert waveform. If desired, an AC-coupled sine wave can be used instead for a more accurate representation.
After data recording is complete, return to the beginning of the B-Alert graph where the stimulus occurred.
Highlight all data before the stimulus, and choose Edit > Clear All.
The MP150 Data is now synched to the B-Alert Data (time point ‘zero’ of the MP150 data file equals time point ‘zero’ of the B-Alert data file).
New Automation Tool Saves Time and Improves Data Consistency
Workflow’s drag-and-drop interface allows you to automate and repeat the signal-conditioning and processing steps your protocol requires in an easy-to-use process without programming or scripting. – Batch process multiple files with confidence that they’re all processed using the same procedure. – Expedite signal conditioning and preparation for analysis with a standardized and automated approach. – Extract measurements, analyze data and get results faster, for meaningful scientific insights. – Save and share workflows with others in your team to help reduce training time and increase collaboration.
AcqKnowledge is an interactive, versatile software program that allows you to manage data in real time. Perform complex data acquisition, stimulation, triggering and analyses using intuitive navigation. Here are some recent AcqKnowledge citations… The nature of vicarious embarrassment. Journal of Business Research, 153, 355-364. Ziegler, A. H., Allen, A. M., Peloza, J., & Norris, J. I. (2022). […]
BIOPAC’s comprehensive Introductory ECG Guide addresses fundamental to advanced concerns to optimize electrocardiography data recording and analysis. Topics include: ECG Complex; Electrical and Mechanical Sequence of a Heartbeat; Systole and Diastole; Configurations for Lead I, Lead II, Lead III, 6-lead ECG, 12-lead ECG, precordial leads; Ventricular Late Potentials (VLPs); ECG Measurement Tools; Automated Analysis Routines for extracting, […]