When using the STMISOC, STMISOD or STMISOE stimulus isolation adapter, it is possible to generate high peak-peak voltages. These voltages are potentially damaging, especially if the high voltage outputs are connected from the stimulator across the heart of the subject–meaning that the heart is potentially in the electrical path from lead to lead. This occurs when stimulation electrodes are placed on opposite sides of the body.
When using the STM200 or STMISOL/LA stimulation units it is similarly possible to generate high voltages. With these latter two units, however, extra precautions are required when using on subjects.
NEVER PLACE STIMULATION ELECTRODES ON OPPPOSITE SIDES OF THE BODY
NEVER PLACE STIMULATION ELECTRODES SO STIMULATION CURRENTS ARE PERMITTED TO FLOW THROUGH THE HEART
This application note details safety guidelines, regulatory standards, and setup for electrical stimulators, including electrial stimulators in the MRI, from BIOPAC.
Offering the industry’s only system that delivers a comprehensive, 24-hour data logging solution in an easy-to-use, easy-to-wear package, the BioNomadix Logger® truly delivers “physiology anywhere.” The BioNomadix Logger is the perfect tool for applications that demand greater degrees of subject freedom and complex experimental design. BioNomadix Loggers wirelessly record physiological data as subjects freely and naturally […]
Solving Physical Stress Issues for Surgeons Physical stress in the operating room is common for surgeons, considering how much time they spend in an upright standing position during surgical procedures. With procedures that require standing for several hours and average reported working times of 60-80 hours a week, surgeons seem to be at risk for musculoskeletal […]
BIOPAC combines eye tracking with mobile data logging and physiology data collection to run experiments in the lab or in the real world BIOPAC Systems, Inc. is pleased to announce new eye tracking integration for researchers who conduct experiments from mobile participants in diverse locations. ETVision (EYE-ETV) eye tracking glasses now integrate with BIOPAC wireless systems and software, […]