This application note lays out some generally recommended methods for recording data from subjects in the MRI or fMRI. BIOPAC Systems, Inc. (BIOPAC) designs systems that can be used to record both biopotential data (such as ECG, EEG and EMG) and transducer data (such as blood pressure, air flow, hand clench strength, finger motion and temperature).
In most MRI configurations, there is a “dual-room” setup. In these cases, there is a “Chamber” room and a “Control” room. The chamber room houses the actual MRI machine and the control room is where the MRI operator sits to manage the MRI scanning sequences.
BIOPAC MRI-related recording equipment is typically setup with the recording equipment data acquisition system and associated amplifiers in the control room (well-away from MRI). Signals are directed, via a filtered cabling system, from the subject—lying in the MRI—to the recording equipment. The cabling system, though conductive, does not contain ferromagnetic materials. The electrodes or transducers attached to the subject in the MRI are plugged into the receiving end of the cabling system.
BSL Home enables students to safely conduct hands-on physiology labs wherever they are! Purchase this complete lab solution to preserve practical lab experience! Includes MP41 Unit+Lead Set with BSL Home software, EL503 electrodes, WRAP1 self-adhering wrap for EEG lessons, and disposable 9 V battery. For instructors—BSL Home makes it easy for instructors to continue running practical […]
BIOPAC provides software and hardware that allows research teams to study physiological measures related to stress. Here are a few notable studies using BIOPAC equipment for ECG, EDA, and EEG/ERP. Moms Moderating Stress Can relaxation techniques help lower stress on both mother and fetus? Using an MP36R Data Acquisition system to record ECG and EDA, this study […]
BIOPAC’s comprehensive Introductory ECG Guide addresses fundamental to advanced concerns to optimize electrocardiography data recording and analysis. Topics include: ECG Complex; Electrical and Mechanical Sequence of a Heartbeat; Systole and Diastole; Configurations for Lead I, Lead II, Lead III, 6-lead ECG, 12-lead ECG, precordial leads; Ventricular Late Potentials (VLPs); ECG Measurement Tools; Automated Analysis Routines for extracting, […]