This application note lays out some generally recommended methods for recording data from subjects in the MRI or fMRI. BIOPAC Systems, Inc. (BIOPAC) designs systems that can be used to record both biopotential data (such as ECG, EEG and EMG) and transducer data (such as blood pressure, air flow, hand clench strength, finger motion and temperature).
In most MRI configurations, there is a “dual-room” setup. In these cases, there is a “Chamber” room and a “Control” room. The chamber room houses the actual MRI machine and the control room is where the MRI operator sits to manage the MRI scanning sequences.
BIOPAC MRI-related recording equipment is typically setup with the recording equipment data acquisition system and associated amplifiers in the control room (well-away from MRI). Signals are directed, via a filtered cabling system, from the subject—lying in the MRI—to the recording equipment. The cabling system, though conductive, does not contain ferromagnetic materials. The electrodes or transducers attached to the subject in the MRI are plugged into the receiving end of the cabling system.
This mini screen-based remote eye tracking bar provides gaze position, dwell time, blink rate, pupil size, and more. Use with an AcqKnowledge Eye Tracking Integration License (ACK100W-EYE) to access valuable eye tracking metrics: Gaze Path, Gaze Plot, Heat Map, and Attention Map. Create user-defined Areas of Interest (AOI) and select Scarf Plot, String Plot, Pie Chart, […]
Solving Physical Stress Issues for Surgeons Physical stress in the operating room is common for surgeons, considering how much time they spend in an upright standing position during surgical procedures. With procedures that require standing for several hours and average reported working times of 60-80 hours a week, surgeons seem to be at risk for musculoskeletal […]
BIOPAC combines eye tracking with mobile data logging and physiology data collection to run experiments in the lab or in the real world BIOPAC Systems, Inc. is pleased to announce new eye tracking integration for researchers who conduct experiments from mobile participants in diverse locations. ETVision (EYE-ETV) eye tracking glasses now integrate with BIOPAC wireless systems and software, […]