This application note is concerned with the possibility of performing high-resolution ECG measurements in the MRI, during EPI. High-resolution ECG recordings are possible in the noisy MRI environment if signal averaging is performed. However, for signal averaging to be performed on ECG data, it’s important to be able to reliably identify locations of QRS complexes on the ECG data record.
By using the BIOPAC ECG100C-MRI amplifier with AcqKnowledgeAnalysis tools, component waves in the ECG data record can be reliably located during EPI. The BIOPAC ECG100C-MRI amplifier can be used to collect electrocardiogram (ECG) data from a subject, in the MRI, during EPI sequencing. The ECG100C-MRI will nearly entirely reject EPI interference to record the ECG manifested at the electrode recording locations. The EPI artifact rejection capability is typically on the order of 40dB, as referenced to the artifact typically present on an ECG waveform not treated with artifact rejection.
The Ventilator Validation Kit supports new prototype development, existing design validation for production testing, mobile field testing and verification, and other uses where accurate pressure and flow measurements are critical. The Ventilator Validation Kit provides ventilator manufacturers and prototype developers with a complete solution for validating new medical/hospital-grade ventilator products intended for human use. The VVK100-SYS includes […]
BIOPAC provides software and hardware that allows research teams to record and analyze numerous body signals for physiological experimentation. Here are a few notable studies using BIOPAC equipment for ECG, EDA, and Respiration data. What Keeps Students More Engaged? The question has been brought to the forefront during COVID, what variations of online learning keep students more engaged? […]
BIOPAC’s comprehensive Introductory ECG Guide addresses fundamental to advanced concerns to optimize electrocardiography data recording and analysis. Topics include: ECG Complex; Electrical and Mechanical Sequence of a Heartbeat; Systole and Diastole; Configurations for Lead I, Lead II, Lead III, 6-lead ECG, 12-lead ECG, precordial leads; Ventricular Late Potentials (VLPs); ECG Measurement Tools; Automated Analysis Routines for extracting, […]