To install, standard Mac practice (mount .dmg, double click the .pkg, follow instructions…will require a computer restart).
2. After computer reboots, verify that the adapter is recognized.
Open System Profiler. Connect ETHUSB. In USB tab, verify that the adapter appears:
3. Go to Software > Extensions and search for AX88772 in the list and verify that version 1.4.0 is installed:
4. Configure the network. Open System Preferences > Network. Select “USB Ethernet” to adjust the adapter settings. Set them to the following manual settings:
Subnet Mask: 255.255.0.0
5. Adjust the network ordering such that USB Ethernet is first on the list.
a) Go to the “gear” at the bottom of the list and choose “Set Service Order” from the list.
b) In the dialog, drag “USB Ethernet” above Ethernet in the list.
c) Click OK and close System Preferences to apply the network settings.
6. Turn on the MP150 (or power cycle it if it was on).
7. Check the amber light on the ETHUSB.
The amber light should turn on when the MP150 is on and may blink periodically but should generally be solid amber.
8. Launch AcqKnowledge.
It should now be possible to establish communications with the MP150.
Additional Steps for Mac OS X 10.10.x – 10.11.x
Make certain all previous steps were followed and the MP150 is connected to the ETHUSB Ethernet adapter.
Disable all Ethernet and WiFi access. (Click the “gear” icon and choose “Make Service Inactive.” This allows the computer to locate the MP150 connected to the Ethernet adapter independently of other MP150s on the network.)
In Mac OS X 10.11.x, make sure a gateway address is present in the USB Ethernet “Router:” field. This is due to a known 10.11.x issue where IP traffic cannot be routed unless the network has a gateway. (Enter 169.254.0.1 into the “Router” field and click “Apply.”)
Launch the AcqKnowledge software.
If desired, re-enable the Ethernet and WiFi access.
If the Ethernet is enabled, the above steps may need to be repeated each time AcqKnowledge is launched. Work-arounds: • Use a direct crossover cable connection from the MP150 to the computer Ethernet port. • Establish communication via the local network without using the crossover cable or ETHUSB adapter
Additional steps for Mac OS X 10.8 – 10.9
In section (2), the driver version will be listed as 1.4.0 if the 10.9 driver is installed.
Add the following intermediate steps:
a) (2.5) Now the USB network interface must be added (10.9+). Open System Preferences > Network. Click on the “+” in the lower left to “Create a new service.
b) In the next dialog that appears, select the “USB Ethernet” interface and give it a default service name. “USB Ethernet” will now be available in the list in the main dialog.
c) (3.5) Unlike previous operating systems, when the cable is connected it will show up as functional with an MP150 when it has a yellow dot.
The Ventilator Validation Kit supports new prototype development, existing design validation for production testing, mobile field testing and verification, and other uses where accurate pressure and flow measurements are critical. The Ventilator Validation Kit provides ventilator manufacturers and prototype developers with a complete solution for validating new medical/hospital-grade ventilator products intended for human use. The VVK100-SYS includes […]
MIT and Harvard researchers developed a new technique to expand ventilator capacity Individualized System for Augmenting Ventilator Efficacy (iSAVE): A Rapidly deployable system to expand ventilator capacity “The iSAVE is a rapidly deployable solution in response to the urgent shortage of mechanical ventilators for respiratory support during the COVID-19 pandemic. In contrast to previously described […]
New Ventilator Validation System Allows Device Manufacturers to Test Ventilator Pressure and Volume System supports rapid development and testing of hardware for COVID-19 treatment Goleta, California – March 24, 2020 BIOPAC Systems, Inc. announced a new Ventilator Validation System for engineering and device manufacturers to use when designing and testing new and existing ventilators for […]