To install, standard Mac practice (mount .dmg, double click the .pkg, follow instructions…will require a computer restart).
2. After computer reboots, verify that the adapter is recognized.
Open System Profiler. Connect ETHUSB. In USB tab, verify that the adapter appears:
3. Go to Software > Extensions and search for AX88772 in the list and verify that version 1.4.0 is installed:
4. Configure the network. Open System Preferences > Network. Select “USB Ethernet” to adjust the adapter settings. Set them to the following manual settings:
Subnet Mask: 255.255.0.0
5. Adjust the network ordering such that USB Ethernet is first on the list.
a) Go to the “gear” at the bottom of the list and choose “Set Service Order” from the list.
b) In the dialog, drag “USB Ethernet” above Ethernet in the list.
c) Click OK and close System Preferences to apply the network settings.
6. Turn on the MP150 (or power cycle it if it was on).
7. Check the amber light on the ETHUSB.
The amber light should turn on when the MP150 is on and may blink periodically but should generally be solid amber.
8. Launch AcqKnowledge.
It should now be possible to establish communications with the MP150.
Additional Steps for Mac OS X 10.10.x – 10.11.x
Make certain all previous steps were followed and the MP150 is connected to the ETHUSB Ethernet adapter.
Disable all Ethernet and WiFi access. (Click the “gear” icon and choose “Make Service Inactive.” This allows the computer to locate the MP150 connected to the Ethernet adapter independently of other MP150s on the network.)
In Mac OS X 10.11.x, make sure a gateway address is present in the USB Ethernet “Router:” field. This is due to a known 10.11.x issue where IP traffic cannot be routed unless the network has a gateway. (Enter 169.254.0.1 into the “Router” field and click “Apply.”)
Launch the AcqKnowledge software.
If desired, re-enable the Ethernet and WiFi access.
If the Ethernet is enabled, the above steps may need to be repeated each time AcqKnowledge is launched. Work-arounds: • Use a direct crossover cable connection from the MP150 to the computer Ethernet port. • Establish communication via the local network without using the crossover cable or ETHUSB adapter
Additional steps for Mac OS X 10.8 – 10.9
In section (2), the driver version will be listed as 1.4.0 if the 10.9 driver is installed.
Add the following intermediate steps:
a) (2.5) Now the USB network interface must be added (10.9+). Open System Preferences > Network. Click on the “+” in the lower left to “Create a new service.
b) In the next dialog that appears, select the “USB Ethernet” interface and give it a default service name. “USB Ethernet” will now be available in the list in the main dialog.
c) (3.5) Unlike previous operating systems, when the cable is connected it will show up as functional with an MP150 when it has a yellow dot.
The Ventilator Validation Kit supports new prototype development, existing design validation for production testing, mobile field testing and verification, and other uses where accurate pressure and flow measurements are critical. The Ventilator Validation Kit provides ventilator manufacturers and prototype developers with a complete solution for validating new medical/hospital-grade ventilator products intended for human use. The VVK100-SYS includes […]
BIOPAC provides software and hardware that allows engineers to study the body. Here are a few notable studies using BIOPAC equipment for Biomedical Engineering research. Rehabilitating the Rehab Industry Could physical therapy be done from the comforts of your own home? If possible, this would open the door for more people with all sorts of ailments to […]
BIOPAC’s comprehensive Introductory ECG Guide addresses fundamental to advanced concerns to optimize electrocardiography data recording and analysis. Topics include: ECG Complex; Electrical and Mechanical Sequence of a Heartbeat; Systole and Diastole; Configurations for Lead I, Lead II, Lead III, 6-lead ECG, 12-lead ECG, precordial leads; Ventricular Late Potentials (VLPs); ECG Measurement Tools; Automated Analysis Routines for extracting, […]