The following is a brief overview of the steps to verify that the USBETH is installed correctly—for more detailed instructions, see the PDF.
Depending on when ETHUSB was purchased, it may have chip AX8872, AX88772A, or AX88772B. BIOPAC is currently shipping ETHUSB with AX88772B. A different adapter or newer ETHUSB may be required for users with chip AX8872 or AX88772A; ASIX has a different page for A but that page doesn’t have a driver released since macOS 11 was released.
1. Download the ASIX Driver that supports macOS 11 for the AX88772B chip (and may also support the AX88772A chip).
After AcqKnowledge has launched and the system has established communication with the MP160, Wi-Fi can be turned back and used during data collection.
4. If turning off Wi-Fi is insufficient, try the additional steps outlined below.
Additional Steps for Mac OS X 10.10.x – 10.11.x
Make certain all previous steps were followed and the MP160/150 is connected to the ETHUSB Ethernet adapter.
Disable all Ethernet and WiFi access. (Click the “gear” icon and choose “Make Service Inactive.” This allows the computer to locate the MP160/150 connected to the Ethernet adapter independently of other MP160s/150s on the network.)
In Mac OS X 10.11.x, make sure a gateway address is present in the USB Ethernet “Router:” field. This is due to a known 10.11.x issue where IP traffic cannot be routed unless the network has a gateway. (Enter 169.254.0.1 into the “Router” field and click “Apply.”)
Launch the AcqKnowledge software.
If desired, re-enable the Ethernet and WiFi access.
If the Ethernet is enabled, the above steps may need to be repeated each time AcqKnowledge is launched. Work-arounds: • Use a direct crossover cable connection from the MP160/150 to the computer Ethernet port. • Establish communication via the local network without using the crossover cable or ETHUSB adapter
Additional steps for Mac OS X 10.8 – 10.9
In section (2), the driver version will be listed as 1.4.0 if the 10.9 driver is installed.
Add the following intermediate steps:
a) (2.5) Now the USB network interface must be added (10.9+). Open System Preferences > Network. Click on the “+” in the lower left to “Create a new service.
b) In the next dialog that appears, select the “USB Ethernet” interface and give it a default service name. “USB Ethernet” will now be available in the list in the main dialog.
c) (3.5) Unlike previous operating systems, when the cable is connected it will show up as functional with an MP160/150 when it has a yellow dot.
BSL Home enables students to safely conduct hands-on physiology labs wherever they are! Purchase this complete lab solution to preserve practical lab experience! Includes MP41 Unit+Lead Set with BSL Home software, EL503 electrodes, WRAP1 self-adhering wrap for EEG lessons, and disposable 9 V battery. For instructors—BSL Home makes it easy for instructors to continue running practical […]
BIOPAC provides software and hardware that allows research teams to study any physiological measure. Here are a few notable studies using BIOPAC equipment for ECG, EDA, and HRV Analysis. Intelligent Interaction Leading to Solutions The research being done in the HCI space is amazing, as scientists continue to make computers more adept at communicating with humans, allowing more […]
BIOPAC’s comprehensive Introductory ECG Guide addresses fundamental to advanced concerns to optimize electrocardiography data recording and analysis. Topics include: ECG Complex; Electrical and Mechanical Sequence of a Heartbeat; Systole and Diastole; Configurations for Lead I, Lead II, Lead III, 6-lead ECG, 12-lead ECG, precordial leads; Ventricular Late Potentials (VLPs); ECG Measurement Tools; Automated Analysis Routines for extracting, […]