This application note is concerned with the nature of ECG100C and ECG100C-MRI timing delays. In particular, the focus is on ECG data collected in fMRI or MRI for the express purpose of triggering magnet scans during imaging. This procedure is important when attempting to image physiological processes in synchronization with the heart.
Typically, ECG data is collected via the ECG100C-MRI Amplifier and this data is piped out of an MP System, to drive a Digital Trigger Unit (DTU300), which in turn drives the magnet scanning process, via magnet scan trigger input. This entire process occurs in real time, not subject to any computer processing delay.
Data demonstrates that is practical to collect ECG data in the fMRI or MRI and use that real-time data to reliably trigger a magnet scan. Although generated magnet artifact does not materially impact the ability of the ECG100C-MRI amplifier’s output to drive the DTU300 (in this data set), note that a synchronized scan will also be completely devoid of artifact before and after the scan period. Accordingly, so long as blanking is employed on the DTU300, any artifact generated in the ECG data will not affect DTU300 triggering operation, despite possibly egregious artifact in the collected ECG data.
This research system ships with an MP160 Data Acquisition with AMI100D Module Interface and AcqKnowledge software licensed for Workflow Automation. MP160 + AMI100D provides a flexible tool for life science researchers. The modular, powerful data acquisition system can be used with BIOPAC’s amplifiers and accessories AND with equipment you already have. To put together a system to […]
Whether using specialized equipment, performing repetitive tasks, or simply going about our daily work routine, the effects of design in our environment as well as the tools we use and the way we use them can impact our health, work performance, and quality of life. Here are a few studies that have recorded and analyzed […]