This application note is concerned with the nature of ECG100C and ECG100C-MRI timing delays. In particular, the focus is on ECG data collected in fMRI or MRI for the express purpose of triggering magnet scans during imaging. This procedure is important when attempting to image physiological processes in synchronization with the heart.
Typically, ECG data is collected via the ECG100C-MRI Amplifier and this data is piped out of an MP System, to drive a Digital Trigger Unit (DTU300), which in turn drives the magnet scanning process, via magnet scan trigger input. This entire process occurs in real time, not subject to any computer processing delay.
Data demonstrates that is practical to collect ECG data in the fMRI or MRI and use that real-time data to reliably trigger a magnet scan. Although generated magnet artifact does not materially impact the ability of the ECG100C-MRI amplifier’s output to drive the DTU300 (in this data set), note that a synchronized scan will also be completely devoid of artifact before and after the scan period. Accordingly, so long as blanking is employed on the DTU300, any artifact generated in the ECG data will not affect DTU300 triggering operation, despite possibly egregious artifact in the collected ECG data.
Electrodermal Activity (EDA) is an objective index for understanding emotional states. Researchers measure EDA to gain insights on psychological or physiological response, leading them down the path of scientific discovery. Because EDA is so important to research, BIOPAC has created tools to optimize event-related EDA analysis. Topics include: how to identify skin conductance responses, convert stim events from E-Prime SuperLab, Presentation, Vizard, etc., into AcqKnowledge events that can used for analysis, and score skin conductance responses using the AcqKnowledge automated EDA analysis tool.
NIRS and CNAP—Pairing great tools for great data Cerebral hemoglobin concentration (blood mass in the brain) is frequently measured in the prefrontal cortex, and systemic arterial blood pressure is known to be a variable factor based on the hemoglobin concentration. A leg pressure cuff was used to constrict and release blood flow to create a […]
BIOPAC’s just released Introductory ECG Guide addresses fundamental to advanced concerns to optimize electrocardiography data recording and analysis. Topics include: ECG Complex; Electrical and Mechanical Sequence of a Heartbeat; Systole and Diastole; Configurations for Lead I, Lead II, Lead III, 6-lead ECG, 12-lead ECG, precordial leads; Ventricular Late Potentials (VLPs); ECG Measurement Tools; Automated Analysis Routines […]