This application note is concerned with the nature of ECG100C and ECG100C-MRI timing delays. In particular, the focus is on ECG data collected in fMRI or MRI for the express purpose of triggering magnet scans during imaging. This procedure is important when attempting to image physiological processes in synchronization with the heart.
Typically, ECG data is collected via the ECG100C-MRI Amplifier and this data is piped out of an MP System, to drive a Digital Trigger Unit (DTU300), which in turn drives the magnet scanning process, via magnet scan trigger input. This entire process occurs in real time, not subject to any computer processing delay.
Data demonstrates that is practical to collect ECG data in the fMRI or MRI and use that real-time data to reliably trigger a magnet scan. Although generated magnet artifact does not materially impact the ability of the ECG100C-MRI amplifier’s output to drive the DTU300 (in this data set), note that a synchronized scan will also be completely devoid of artifact before and after the scan period. Accordingly, so long as blanking is employed on the DTU300, any artifact generated in the ECG data will not affect DTU300 triggering operation, despite possibly egregious artifact in the collected ECG data.
MP160 data acquisition and analysis systems with AcqKnowledge software provide a flexible tool for life science research. MP160 data acquisition workstations include all hardware, software, cables, and guides you need to get started. MP160 Data Acquisition Hardware—Ethernet-ready data acquisition and analysis unit can record multiple channels with differing sample rates; record at speeds up to 400 kHz (aggregate). […]
BIOPAC provides software and hardware that allows research teams to study stimulus presentation data. Here are a few notable studies using BIOPAC equipment for research on stimuli’s effects using ECG and SCL data. Your Face Scares Me This study addressed the impact of individual differences in social anxiety by examining the effects of perceptual load and stimulus valence when […]
BIOPAC’s comprehensive Introductory ECG Guide addresses fundamental to advanced concerns to optimize electrocardiography data recording and analysis. Topics include: ECG Complex; Electrical and Mechanical Sequence of a Heartbeat; Systole and Diastole; Configurations for Lead I, Lead II, Lead III, 6-lead ECG, 12-lead ECG, precordial leads; Ventricular Late Potentials (VLPs); ECG Measurement Tools; Automated Analysis Routines for extracting, […]