This application note is concerned with the nature of ECG100C and ECG100C-MRI timing delays. In particular, the focus is on ECG data collected in fMRI or MRI for the express purpose of triggering magnet scans during imaging. This procedure is important when attempting to image physiological processes in synchronization with the heart.
Typically, ECG data is collected via the ECG100C-MRI Amplifier and this data is piped out of an MP System, to drive a Digital Trigger Unit (DTU300), which in turn drives the magnet scanning process, via magnet scan trigger input. This entire process occurs in real time, not subject to any computer processing delay.
Data demonstrates that is practical to collect ECG data in the fMRI or MRI and use that real-time data to reliably trigger a magnet scan. Although generated magnet artifact does not materially impact the ability of the ECG100C-MRI amplifier’s output to drive the DTU300 (in this data set), note that a synchronized scan will also be completely devoid of artifact before and after the scan period. Accordingly, so long as blanking is employed on the DTU300, any artifact generated in the ECG data will not affect DTU300 triggering operation, despite possibly egregious artifact in the collected ECG data.
Basic principles of physiological data collection covers fundamentals of data collection and equipment configuration equipment to help you refine experiment protocols and avoid costly missteps. Frazer Findlay, CEO of BIOPAC, discusses common mistakes with physiology recording and shares his secrets for collecting great data. Topics include: How to prep a subject and where to place electrodes; Which type of electrodes work for different body signals; Analog to Digital Conversion; Sample rates (correct vs. incorrect); Filtering; and Scaling and calibration.Watch On Demand Now!
Recent psychophysiological studies feature BIOPAC’s MP Research series of Data Recorders and AcqKnowledge software. Creating Adaptive Assistants for helicopter crews based on Real-time Physiological Data In the Mercedes S-Class, biometric tools track a driver’s eyes, heart rate, and other physiological data, with one goal in mind: Making sure drivers are alert and safe on the […]
BIOPAC’s just released Introductory ECG Guide addresses fundamental to advanced concerns to optimize electrocardiography data recording and analysis. Topics include: ECG Complex; Electrical and Mechanical Sequence of a Heartbeat; Systole and Diastole; Configurations for Lead I, Lead II, Lead III, 6-lead ECG, 12-lead ECG, precordial leads; Ventricular Late Potentials (VLPs); ECG Measurement Tools; Automated Analysis Routines […]