This application note is concerned with the nature of ECG100C and ECG100C-MRI timing delays. In particular, the focus is on ECG data collected in fMRI or MRI for the express purpose of triggering magnet scans during imaging. This procedure is important when attempting to image physiological processes in synchronization with the heart.
Typically, ECG data is collected via the ECG100C-MRI Amplifier and this data is piped out of an MP System, to drive a Digital Trigger Unit (DTU300), which in turn drives the magnet scanning process, via magnet scan trigger input. This entire process occurs in real time, not subject to any computer processing delay.
Data demonstrates that is practical to collect ECG data in the fMRI or MRI and use that real-time data to reliably trigger a magnet scan. Although generated magnet artifact does not materially impact the ability of the ECG100C-MRI amplifier’s output to drive the DTU300 (in this data set), note that a synchronized scan will also be completely devoid of artifact before and after the scan period. Accordingly, so long as blanking is employed on the DTU300, any artifact generated in the ECG data will not affect DTU300 triggering operation, despite possibly egregious artifact in the collected ECG data.
The ETVision (ETV) wearable unit resembles an eyeglasses frame that can be worn by itself or over the participant’s prescription eyeglasses. It contains miniature “eye cameras” that view each eye and a “scene camera” that views the scene in front of the participant, and a microphone. An optional visor can be mounted to the front […]
Solving Physical Stress Issues for Surgeons Physical stress in the operating room is common for surgeons, considering how much time they spend in an upright standing position during surgical procedures. With procedures that require standing for several hours and average reported working times of 60-80 hours a week, surgeons seem to be at risk for musculoskeletal […]
BIOPAC combines eye tracking with mobile data logging and physiology data collection to run experiments in the lab or in the real world BIOPAC Systems, Inc. is pleased to announce new eye tracking integration for researchers who conduct experiments from mobile participants in diverse locations. ETVision (EYE-ETV) eye tracking glasses now integrate with BIOPAC wireless systems and software, […]