The Upgrade Procedure for BSL 2 to BSL 3 assumes you have a BIOPAC ISA100A card installed in your computer that has been successfully running the MP30 acquisition unit and requires that you know the I/O address for your ISA100A card.
The documented procedures will help you determine the I/O address for your ISA100A card. You need to determine the address prior to running the upgrade installation. The Upgrade Procedure follows the address determination options.
The fEMG Electromyography Smart Amplifier is designed specifically for recording facial EMG signals and electrical activity from other small muscle groups. Use with AcqKnowledge™ software to analyze facial expressions and startle paradigms. BIOPAC’s new Smart Amplifiers are designed for great data. Smart Amplifiers improve performance by amplifying the physiological signal close to the subject, which allows a high-level […]
A System to Measure Physiological Response During Social Interaction in VR Children with Autism Spectrum Disorder (ASD) have a difficult experience in responding to social cues and actions. This pilot study measured the effectiveness of Virtual Reality in creating a social environment for children, measured with a BIOPAC MP Series data acquisition system and Worldviz […]
BIOPAC’s just released Introductory ECG Guide addresses fundamental to advanced concerns to optimize electrocardiography data recording and analysis. Topics include: ECG Complex; Electrical and Mechanical Sequence of a Heartbeat; Systole and Diastole; Configurations for Lead I, Lead II, Lead III, 6-lead ECG, 12-lead ECG, precordial leads; Ventricular Late Potentials (VLPs); ECG Measurement Tools; Automated Analysis Routines […]