When on-line averaging is invoked, some signal amplitude excursions may need to be excluded during the averaging process. Artifact Rejection is a feature for just this purpose. When artifact rejection is enabled, the MP150 System will ignore any trials that contain signals exceeding the artifact rejection thresholds, keep track of how many trials have been rejected, and add that number of trials to the total number of trials to be acquired. This allows users to “re-try” a trial that was rejected due to the presence of artifact.
Occasionally during an acquisition, extreme levels of artifact will be present for one reason or another. Checking artifact rejection allows users to determine what signal levels constitute artifact, and have the MP150 System reject these trials.
This application note provides a procedure/test case to confirm that AcqKnowledgeArtifact Rejection is functioning properly.
MedelOpt® Infinity systems provide high-density NIRS with EEG for very high density whole brain mapping and custom advanced montages with variable depths. The flexible, adaptable headset allows acquisition for up to 8 hours. MedelOpt adjusts to real-world situations or Virtual Reality environments with a specific design to fit any head mounted VR display. Use for research […]
The autonomic nervous system (ANS) is responsible for regulating the functions of all the body’s organs. Take a look at this month’s citations to see how to apply BIOPAC’s tools and learn techniques for ANS research… Comparable responses to a wide range of olfactory stimulation in women and men. Scientific Reports, 13(1), 9059. Lillqvist, M., Claeson, […]
BIOPAC’s comprehensive Introductory ECG Guide addresses fundamental to advanced concerns to optimize electrocardiography data recording and analysis. Topics include: ECG Complex; Electrical and Mechanical Sequence of a Heartbeat; Systole and Diastole; Configurations for Lead I, Lead II, Lead III, 6-lead ECG, 12-lead ECG, precordial leads; Ventricular Late Potentials (VLPs); ECG Measurement Tools; Automated Analysis Routines for extracting, […]