When on-line averaging is invoked, some signal amplitude excursions may need to be excluded during the averaging process. Artifact Rejection is a feature for just this purpose. When artifact rejection is enabled, the MP150 System will ignore any trials that contain signals exceeding the artifact rejection thresholds, keep track of how many trials have been rejected, and add that number of trials to the total number of trials to be acquired. This allows users to “re-try” a trial that was rejected due to the presence of artifact.
Occasionally during an acquisition, extreme levels of artifact will be present for one reason or another. Checking artifact rejection allows users to determine what signal levels constitute artifact, and have the MP150 System reject these trials.
This application note provides a procedure/test case to confirm that AcqKnowledgeArtifact Rejection is functioning properly.
Easily record great electrodermal activity (EDA, GSR) data. BIOPAC’s new Smart Amplifiers are designed for great data. Smart Amplifiers improve performance by amplifying the physiological signal close to the subject, which allows a high-level voltage connection to the data acquisition system and reduces noise artifact. AcqKnowledge Smart Amp setup includes guides and prompts to prevent […]
BIOPAC provides software and hardware that allows research teams to record and analyze numerous body signals for physiological experimentation. Here are a few notable studies using BIOPAC equipment for ECG, EDA, and Respiration data. What Keeps Students More Engaged? The question has been brought to the forefront during COVID, what variations of online learning keep students more engaged? […]
BIOPAC’s comprehensive Introductory ECG Guide addresses fundamental to advanced concerns to optimize electrocardiography data recording and analysis. Topics include: ECG Complex; Electrical and Mechanical Sequence of a Heartbeat; Systole and Diastole; Configurations for Lead I, Lead II, Lead III, 6-lead ECG, 12-lead ECG, precordial leads; Ventricular Late Potentials (VLPs); ECG Measurement Tools; Automated Analysis Routines for extracting, […]