fNIR Systems extract measurements from an LED light source at wavelengths of 730 nm and 850 nm, and also provide a unique feature that can be useful for identifying certain types of noise within recordings. This feature enables the fNIR imager to record measurements under “dark current” conditions.
Dark current measurements are always recorded in fNIR data output files (.nir and .oxy) and can be monitored during sensor placement, baseline, and overall testing measurement periods by selecting **** under **** buttons in COBI studio.
Time periods during which dark current measurements do not remain flat in close correlation with measurements taken at 730 nm and 850 nm can help identify noisy data segments. These regions can subsequently be marked and discarded from further hemoglobin content extraction analysis.
Noise is not always as obvious the sample shown, and dark current measurements offer a justifiable secondary measure for verifying the presence of noise in all cases.
The fEMG Electromyography Smart Amplifier is designed specifically for recording facial EMG signals and electrical activity from other small muscle groups. Use with AcqKnowledge™ software to analyze facial expressions and startle paradigms. BIOPAC’s new Smart Amplifiers are designed for great data. Smart Amplifiers improve performance by amplifying the physiological signal close to the subject, which allows a high-level […]
Recent Exercise Physiology studies feature BIOPAC’s MP Research series of Data Collection recorders and AcqKnowledge software. These studies promote healthy living and observe muscle response to normal physiological motions during exercise! Trampoline Workouts for Seniors? As we get older, our joints ache and muscles stiffen—but scientists are finding new ways to not only help us […]
BIOPAC’s just released Introductory ECG Guide addresses fundamental to advanced concerns to optimize electrocardiography data recording and analysis. Topics include: ECG Complex; Electrical and Mechanical Sequence of a Heartbeat; Systole and Diastole; Configurations for Lead I, Lead II, Lead III, 6-lead ECG, 12-lead ECG, precordial leads; Ventricular Late Potentials (VLPs); ECG Measurement Tools; Automated Analysis Routines […]