fNIR Systems extract measurements from an LED light source at wavelengths of 730 nm and 850 nm, and also provide a unique feature that can be useful for identifying certain types of noise within recordings. This feature enables the fNIR imager to record measurements under “dark current” conditions.
Dark current measurements are always recorded in fNIR data output files (.nir and .oxy) and can be monitored during sensor placement, baseline, and overall testing measurement periods by selecting **** under **** buttons in COBI studio.
Time periods during which dark current measurements do not remain flat in close correlation with measurements taken at 730 nm and 850 nm can help identify noisy data segments. These regions can subsequently be marked and discarded from further hemoglobin content extraction analysis.
Noise is not always as obvious the sample shown, and dark current measurements offer a justifiable secondary measure for verifying the presence of noise in all cases.
MedelOpt® Mobility systems provide high-density NIRS with EEG available in a flexible, adaptable headset for high density mapping with complete mobility with an unlimited range of acquisition for up to 8 hours. MedelOpt Mobility adjusts to real-world situations or virtual reality (VR) environments with a specific design to fit any head mounted VR display. Use for […]View All
Discovering and understanding what motivates humans to produce better results has intrigued many researchers. Many researchers have explored the psychophysiological processes that drive our behavior. Here are some recent studies that have used BIOPAC systems to research motivation… Motivation and Pleasure Deficits Undermine the Benefits of Social Affiliation in Psychosis. Blanchard, J. J., Smith, J. […]
BIOPAC’s comprehensive Introductory ECG Guide addresses fundamental to advanced concerns to optimize electrocardiography data recording and analysis. Topics include: ECG Complex; Electrical and Mechanical Sequence of a Heartbeat; Systole and Diastole; Configurations for Lead I, Lead II, Lead III, 6-lead ECG, 12-lead ECG, precordial leads; Ventricular Late Potentials (VLPs); ECG Measurement Tools; Automated Analysis Routines for extracting, […]Read All