fNIR Systems extract measurements from an LED light source at wavelengths of 730 nm and 850 nm, and also provide a unique feature that can be useful for identifying certain types of noise within recordings. This feature enables the fNIR imager to record measurements under “dark current” conditions.
Dark current measurements are always recorded in fNIR data output files (.nir and .oxy) and can be monitored during sensor placement, baseline, and overall testing measurement periods by selecting **** under **** buttons in COBI studio.
Time periods during which dark current measurements do not remain flat in close correlation with measurements taken at 730 nm and 850 nm can help identify noisy data segments. These regions can subsequently be marked and discarded from further hemoglobin content extraction analysis.
Noise is not always as obvious the sample shown, and dark current measurements offer a justifiable secondary measure for verifying the presence of noise in all cases.
Smart Center Stand-alone, Portable Wireless Physiology Systems Smart Center Essentials entry-level system includes a Smart Center Device (BN-SMART with USB power cable), 2 or 3 Transmitters (BN-xxx-T with chargers); AcqKnowledge for Smart Center, and a case. Smart Center Enhanced adds a Logger (BN-LOGGER) and Basic Scripting License (ACK100W-BAS) to the Smart Center Device (BN-SMART with powercable), 3 […]
Recent physiological studies feature BIOPAC’s Wireless BioNomadix Data Recorders Using Metronomes and Rhythmic stimulation to help people with Parkinson’s disease walk For those living with Parkinson’s disease, even simple tasks can take too long to accomplish properly, such as crossing a room or picking up an object. In this study, participants with diagnosed cases of […]
BIOPAC’s just released Introductory ECG Guide addresses fundamental to advanced concerns to optimize electrocardiography data recording and analysis. Topics include: ECG Complex; Electrical and Mechanical Sequence of a Heartbeat; Systole and Diastole; Configurations for Lead I, Lead II, Lead III, 6-lead ECG, 12-lead ECG, precordial leads; Ventricular Late Potentials (VLPs); ECG Measurement Tools; Automated Analysis Routines […]