B-Alert headsets are shipped paired to either a B-Alert Dongle or an ESU. If the pairing is lost, the devices must be re-synced using the B-Alert Live software, which is included with your AcqKnowledge installation. B-Alert Live supports AcqKnowledge B-Alert functionality and is installed to the following default directory: Main Drive:\ABM\B-Alert\Rome\
Browse to the above directory and launch the B-Alert Live software by double-clicking the Rome.exe icon.
Plug the Bluetooth receiver (i.e., dongle or ESU) into your PC. Note: Ensure that all nearby B-Alert products, including headsets and other dongles/ESUs, are turned off and unplugged.
Switch ON the B-Alert headset. The Red LED will illuminate for the first 3-4 seconds, followed by 10 seconds of Green LED flashing. Wait until the headset LED pattern is Flashing Green and Solid Red before proceeding to the next step:
In the B-Alert Live software’s Operations menu, select “Sync Devices.”
Click the “Search Devices” button and wait. Allow 45-60 seconds for the USB device to search for other Bluetooth devices in range.
After the search has completed, all Bluetooth devices identified in range will appear under the “Select Device” drop down menu.
Look for the B-Alert headset. The device should have a 12 digit BDA address followed by: (ABM EEG XXXXXXXXX). Example: 000BCE042CCA (ABM EEG 300103056):
Click “Connect” and wait for the pop-up window indicating that the headset has connected: “Device connection succeeded. Please restart device.”
Click “OK” and then switch OFF the Headset.
Unplug the ESU receiver, and then re-plug it into the laptop/computer.
Switch the headset back ON; wait 5-10 seconds for the Green LED on headset to turn Solid Green, signaling the headset has synced.
IMPORTANT: During pairing, the distance between the headset and Bluetooth receiver should be line-of-sight and less than 3 meters.
BSL Home rentals enable students to conduct hands-on physiology labs in the safety of their own home, dorm room, or any remote location while social distancing and stay-at-home orders keep them out of campus labs Complete lab solution to preserve practical lab experience! For instructors—BSL Home makes it easy for instructors to continue running practical […]
Comparing Diaphragm Tissue Between Healthy and Ill Patients The human diaphragm is one of the primary muscles used in respiration, contracting and expanding to control breathing. To measure how much diaphragms move for critically ill patients (in an Intensive Care Unit) compared to diaphragms in a healthy subject, patients at “Papageorgiou” General Hospital in Thessaloniki, […]
BIOPAC’s ventilator testing equipment and expertise supports STARK Industries’ effort to manufacture, validate, and test NASA JPL’s VITAL Ventilators Goleta, California – June 16, 2020 BIOPAC Systems, Inc. announced a partnership with STARK Industries, LLC to provide ventilator validation and testing equipment for the mass production of ventilators. STARK Industries was awarded a global license […]