B-Alert headsets are shipped paired to either a B-Alert Dongle or an ESU. If the pairing is lost, the devices must be re-synced using the B-Alert Live software, which is included with your AcqKnowledge installation. B-Alert Live supports AcqKnowledge B-Alert functionality and is installed to the following default directory: Main Drive:\ABM\B-Alert\Rome\
Browse to the above directory and launch the B-Alert Live software by double-clicking the Rome.exe icon.
Plug the Bluetooth receiver (i.e., dongle or ESU) into your PC. Note: Ensure that all nearby B-Alert products, including headsets and other dongles/ESUs, are turned off and unplugged.
Switch ON the B-Alert headset. The Red LED will illuminate for the first 3-4 seconds, followed by 10 seconds of Green LED flashing. Wait until the headset LED pattern is Flashing Green and Solid Red before proceeding to the next step:
In the B-Alert Live software’s Operations menu, select “Sync Devices.”
Click the “Search Devices” button and wait. Allow 45-60 seconds for the USB device to search for other Bluetooth devices in range.
After the search has completed, all Bluetooth devices identified in range will appear under the “Select Device” drop down menu.
Look for the B-Alert headset. The device should have a 12 digit BDA address followed by: (ABM EEG XXXXXXXXX). Example: 000BCE042CCA (ABM EEG 300103056):
Click “Connect” and wait for the pop-up window indicating that the headset has connected: “Device connection succeeded. Please restart device.”
Click “OK” and then switch OFF the Headset.
Unplug the ESU receiver, and then re-plug it into the laptop/computer.
Switch the headset back ON; wait 5-10 seconds for the Green LED on headset to turn Solid Green, signaling the headset has synced.
IMPORTANT: During pairing, the distance between the headset and Bluetooth receiver should be line-of-sight and less than 3 meters.
The ETVision (ETV) wearable unit resembles an eyeglasses frame that can be worn by itself or over the participant’s prescription eyeglasses. It contains miniature “eye cameras” that view each eye and a “scene camera” that views the scene in front of the participant, and a microphone. An optional visor can be mounted to the front […]
Solving Physical Stress Issues for Surgeons Physical stress in the operating room is common for surgeons, considering how much time they spend in an upright standing position during surgical procedures. With procedures that require standing for several hours and average reported working times of 60-80 hours a week, surgeons seem to be at risk for musculoskeletal […]
BIOPAC combines eye tracking with mobile data logging and physiology data collection to run experiments in the lab or in the real world BIOPAC Systems, Inc. is pleased to announce new eye tracking integration for researchers who conduct experiments from mobile participants in diverse locations. ETVision (EYE-ETV) eye tracking glasses now integrate with BIOPAC wireless systems and software, […]