B-Alert headsets are shipped paired to either a B-Alert Dongle or an ESU. If the pairing is lost, the devices must be re-synced using the B-Alert Live software, which is included with your AcqKnowledge installation. B-Alert Live supports AcqKnowledge B-Alert functionality and is installed to the following default directory: Main Drive:\ABM\B-Alert\Rome\
Browse to the above directory and launch the B-Alert Live software by double-clicking the Rome.exe icon.
Plug the Bluetooth receiver (i.e., dongle or ESU) into your PC. Note: Ensure that all nearby B-Alert products, including headsets and other dongles/ESUs, are turned off and unplugged.
Switch ON the B-Alert headset. The Red LED will illuminate for the first 3-4 seconds, followed by 10 seconds of Green LED flashing. Wait until the headset LED pattern is Flashing Green and Solid Red before proceeding to the next step:
In the B-Alert Live software’s Operations menu, select “Sync Devices.”
Click the “Search Devices” button and wait. Allow 45-60 seconds for the USB device to search for other Bluetooth devices in range.
After the search has completed, all Bluetooth devices identified in range will appear under the “Select Device” drop down menu.
Look for the B-Alert headset. The device should have a 12 digit BDA address followed by: (ABM EEG XXXXXXXXX). Example: 000BCE042CCA (ABM EEG 300103056):
Click “Connect” and wait for the pop-up window indicating that the headset has connected: “Device connection succeeded. Please restart device.”
Click “OK” and then switch OFF the Headset.
Unplug the ESU receiver, and then re-plug it into the laptop/computer.
Switch the headset back ON; wait 5-10 seconds for the Green LED on headset to turn Solid Green, signaling the headset has synced.
IMPORTANT: During pairing, the distance between the headset and Bluetooth receiver should be line-of-sight and less than 3 meters.
In this online webinar, Frazer Findlay, CEO of BIOPAC, will explain how to set up a Facial EMG study and record great Facial EMG data using the newest Smart Amplifiers. He will demonstrate where to place the electrodes and best practices for recording signals from the muscles tied to our emotions. Join us Thursday, April 25th 8:00 AM to 9:00 PM PT
Recent Studies using BIOPACs MP-Series (MP160, MP150) data recorders, ECG Amplifiers, and BioNomadix Wireless recorders: Singing style to infants Mothers around the world sing to infants, presumably to regulate their mood and arousal. Lullabies and playsongs differ stylistically and have distinctive goals. Mothers sing lullabies to soothe and calm infants, and playsongs to engage and […]
Alex Dimov, BIOPAC Sales Executive, was interviewed at the Neuromarketing World Forum by Inside Marketing to discuss how virtual reality can improve marketing research. Alex explains how VR can be used to present virtual iterations of a product to save time and money, and notes that measuring physiological responses adds lots of information for product and messaging […]