This Application Note outlines the precise test equipment calibration required to ensure proper signal-to-noise levels (S/N) and signal time delays for Auditory Brainstem Response (ABR) measurements using an MP150 data acquisition system and AcqKnowledge 4.1 or above software.
Whenever a sound wave is detected by the ear, it triggers a number of neuro-physiological responses along the auditory pathway. An action potential is conducted along the eighth nerve, the brainstem, and finally to the brain.
About one third of a second after initial stimulation, the signal evokes a response in the area of the brain where sounds are interpreted. These response signals are so small that they are frequently masked by the “background noise” of electrical activity in the brain. This interference is random with respect to the initial stimulation.
If you find an average of this interference signal, it will approach zero. However, since the evoked response is dependent on the stimulus, it should occur at the same time relative to the stimulus when the process is repeated. Therefore, if you repeat this stimulus-response process many times the “background noise” will disappear and what will be left is the neuro-physiological response of principal interest.
For ABR studies, calibrate the OUT101 Tubephone with anEtymotic ER-7C Probe Microphone—this microphone provides a calibrated output voltage which is a function of applied Sound Pressure Level (SPL). The sensitivity is 50 mV/Pascal (-46 dB re: 1 V/uBar): 0 dB SPL = 0 dBuV. Place the Probe Microphone insert tube in the auditory canal prior to the insertion of the OUT101 foam tip. The OUT101 Tubephone sound delivery tube and the Probe Microphone sound input tube will then be exposed to the same auditory chamber. Accordingly, the SPL is recorded, via the Probe Microphone, simultaneously with applied auditory stimulus from the OUT101 Tubephone.
Simple, portable, accurate, wireless This complete system provides acquisition and analysis of high fidelity wireless EEG plus ECG with the B-Alert X10 headset and sensor strips (1 small, 1 medium), plus AcqKnowledge software with powerful analysis tools, including automated scoring and reporting options, and B-Alert Cognitive State analysis software. The B-Alert X10 mobile-wireless EEG system delivers real-time measurements for a […]
Brain Connectivity and Human Responses to Music: While it has been proven that people find music aesthetically appealing because of neural circuitry, it is unclear why individuals received different levels of enjoyment to different types of music. Researchers from Wesleyan University used a combination of survey data and behavioral and psychophysiological measures to try and understand […]
Virtual Reality Muscle Training in Stroke Patients: Muscle weakening of the lower extremities has a significant correlation with loss of balance and walking ability in stroke patients. Researchers at Sehan University studied the effect of a virtual reality-based eccentric training program on lower extremity muscle activation and balance for rehabilitation of stroke patients. BIOPAC’s EMG100C surface […]