This Application Note outlines the precise test equipment calibration required to ensure proper signal-to-noise levels (S/N) and signal time delays for Auditory Brainstem Response (ABR) measurements using an MP150 data acquisition system and AcqKnowledge 4.1 or above software.
Whenever a sound wave is detected by the ear, it triggers a number of neuro-physiological responses along the auditory pathway. An action potential is conducted along the eighth nerve, the brainstem, and finally to the brain.
About one third of a second after initial stimulation, the signal evokes a response in the area of the brain where sounds are interpreted. These response signals are so small that they are frequently masked by the “background noise” of electrical activity in the brain. This interference is random with respect to the initial stimulation.
If you find an average of this interference signal, it will approach zero. However, since the evoked response is dependent on the stimulus, it should occur at the same time relative to the stimulus when the process is repeated. Therefore, if you repeat this stimulus-response process many times the “background noise” will disappear and what will be left is the neuro-physiological response of principal interest.
For ABR studies, calibrate the OUT101 Tubephone with anEtymotic ER-7C Probe Microphone—this microphone provides a calibrated output voltage which is a function of applied Sound Pressure Level (SPL). The sensitivity is 50 mV/Pascal (-46 dB re: 1 V/uBar): 0 dB SPL = 0 dBuV. Place the Probe Microphone insert tube in the auditory canal prior to the insertion of the OUT101 foam tip. The OUT101 Tubephone sound delivery tube and the Probe Microphone sound input tube will then be exposed to the same auditory chamber. Accordingly, the SPL is recorded, via the Probe Microphone, simultaneously with applied auditory stimulus from the OUT101 Tubephone.
For reduced application time and increased subject comfort, use the BioNomadix EEG Cap System to record multiple EEG channels. The medium Lycra stretch cap (54-58cm) holds 19 embedded tin electrodes closely to the subject’s head. Electrodes are pre-positioned in the international 10/20 montage, so even novice EEG researchers can minimize electrode placement errors.
Humans, much like animals, separate themselves into different social hierarchies often associated by some specific sort of status signal. While animals may assign hierarchies strictly through physiological measures, the Biosocial Model states that speech is often the strongest form of status signaling for humans. BIOPAC user Allan Mazur recently was part of the research team…
In Technology: Photoplethysmography with a Smart Phone Pulse transit time (PTT) recordings are widely used to make inferences about dynamics of vascular walls and blood pressure variations, parameters useful to the diagnosis of several respiratory conditions. Chinese researchers have developed a method for noninvasive photoplethysmograhic imaging (PPGi) using the dual cameras of a common smartphone,…